1. Output vibration signal error:
Internal source signal：5, 10, 20, 40, 80, 160, 325, 495, 995Hz ≤1%
Frequency accuracy：5, 10, 20, 40, 80, 160, 325, 495, 995Hz ≤3%
2. The maximum vibration amplitude and maximum load:
Because the WS-JX-3 type vibration calibration table system uses a shaker body that is small, the amplitude of the calibrator is not the same at different frequencies and at the sensor’s different weights. The maximum vibration amplitude of acceleration, velocity, and displacement and the maximum mean load the sensor is calibrated at are at a certain frequency and a particular weight.
3. Adjustable frequency:
5Hz, 10Hz, 20Hz, 40Hz, 80Hz, 160Hz, 325Hz, 495Hz, 995Hz
Because the weight of the shaker body of the JX-3C sensor calibrator is less. The amplitude of its shaker is different when calibrating with different weights of sensors. The following list is the max amplitude of acceleration, velocity, and displacement at different frequencies and different ranges of the weight of the sensor to be calibrated.
|Weight of Sensor≤300g||Weight of Sensor≤600g|
[Note] “ * ” means the displacement is less than 1μm.
Panel Diagram and Function